|Place of Origin:||China|
|Certification:||CE、 FDA 、CFDA、KFDA|
|Supply Ability:||10 working days after received your payment|
|Density Before Sintering (g/cm3):||3.1±0.5 G/cm3||Density After Sintering (g/cm3):||6.08±0.01 G/cm3|
|CTE After Sintering (25-200℃):||(10.5±1.0)x10-6K-1||3-point Bending Strength After Sintering (MPa):||1100Mpa|
|Average Transmittance:||25~39%||Chemical Solubility After Sintering (μg/cm2):||< 100|
|Radioactivity After Sintering (Bq/g):||<0.1||Sintering Temperature (℃):||1450-1550℃Recommended 1530℃|
Dental Laboratory Milling Supplies Core Dental Ceramics ST 16 Pre - shaded
ST COLOR block: suitable for single shaded crown and bridge with VITA classic 16 shades(A1/A2/A3/A3.5/A4/B1/B2/B3/B4/C1/C2/C3/C4/D2/D3/D4)
3-4 units bridge
5-6 units bridge
Bridges over 7 units
1 .Place a zirconia block in your CAD/CAM machine and machine it following the manufacturer’s instructions.
2 .After machining is completed, remove the zirconia block from the CAD/CAM machine first. Then remove the framework from the zirconia block with a diamond disc. Subsequently, place the framework on a sintering tray with zirconia beads and put them in the oven for sintering.
3 .Program the furnace according to its instruction manual and zirconia block’s sintering chart, and then start sintering.
4 .Do not remove the sintered framework until it is fully cooled down. Examine the sintered framework for wall thickness, margin fit and seat fit. If necessary, make minor adjustments with a water cooled diamond tool.
5 .Make sure no wall thickness is less than 0.5mm and no connector is thinner than 9mm2.
6 .Rinse the sintered framework with water and dry completely.